- Ward round
This innovative product “Digital Hand-held diagnostic set” is Class-II (ophthalmoscope)/Class-I (otoscope and dermoscope) Medical Device.
It is expected to substitute for the traditional ophthalmoscope, otoscope and dermoscope by the use of the digital photographic solution. This medical device is provided to capture the digital photograph or video of eye-fundus, ear canal and tympanic membrane, epidermis and dermis of skin. Following the global trend of electronic medical records and telehealthcare networks, this people-oriented medical imaging product will be widely applied in doctors’ offices, clinics, skilled nursing facilities and healthcare station.
|Image Resolution||1920 x 1080 pixels|
|LCD Monitor||3.5″ Full ColorTFT-LCD|
|Interface||Mini USB, AV out port|
|File Transfer||Minis USB Port to PC|
|Dynamic Video Output||AV output／USB|
|File Storage||Supports 2G to 32GB|
|Rechargeable Lithium Battery|
|External Power||Power 100~240VAC 50／60Hz|
|Operating Time||3 hours|
|Charging Time||5 hours|
Q: How to adjust the position of lens when mounted horus scope to slit lamp?
A: First, align the center of lens with the mark of eye height, and then, it’s just the same as operating a slit lamp for image capture.
Q: I had disabled standby mode, but the control unit still switched to standby mode
A: Standby Disable setting returns to default value after re-boot the machine
Q: Orange light and blue light flashing alternatively while charging, but the control unit is still low on battery even after charging for hours
A: Please check if the battery is installed in the control unit.